Follow the steps below to replace PCA modules in the Calibrator. Refer to Figure 4-3 for a disassembly illustration and the location of the PCA modules.
1. Turn the Calibrator off and disconnect line power.
2. Remove the six screws holding the top cover in place. Remove the top cover.
3. Remove the two screws holding the guard cover in place. Tilt the guard cover up and leave vertical or remove the guard cover.
4. Remove the required module using the module extractors.
Figure 4-3. Exploded View of the Calibrator apv102f.eps
Figure 4-3. Exploded View of the Calibrator
Was this article helpful?